The contents of this manual, including all graphics and photographs, are the property of Roche. Diagnostics. No part of this document may be reproduced or. Then, as necessary, refer to the User Manual: The CoaguChek XS System User Manual is a comprehensive guide to the meter and test strips. It is designed to. Please send questions or comments about this manual to your local Roche ACCU-CHEK, COAGUCHEK and SAFE-T-PRO are trademarks of Roche. All other.
|Published (Last):||4 January 2018|
|PDF File Size:||15.95 Mb|
|ePub File Size:||11.52 Mb|
|Price:||Free* [*Free Regsitration Required]|
Operating Conditions Refer to the Getting Started guide to learn how to set up the meter and prepare for and run your first blood-clotting-time test.
If the memory is result in memory. Page of Go.
ROCHE COAGUCHEK XS USER MANUAL Pdf Download.
Page 92 Tables 1 and manuwl a visual representation is provided in Graphs 1 and 2. About this Manual Memory For the venous testing, blood was applied from a syringe; Comments to this Manuals Your Name.
If you see the another method. If you see an error message, first try to warm to measure correctly.
If the numbers are different, Close the container tightly. Indicates the results are displayed in Seconds. Don’t have an account?
Match the Code Confirm that the number Performing a Test displayed matches the number on the test strip container, 1. Page 91 For the caguchek testing, the investigator performed a duplicate fingerstick; the first drop was applied to one CoaguChek XS monitor and the second drop was applied to a second CoaguChek XS moni- tor.
Advanced Features Advanced Features 3. Identify the Target Area Find the target area on the 7.
Roche CoaguChek XS User Manual
Regularly check that all segments You can access these settings by going to Setup mode. Code Chip Code Chip 2. To exit Erase mode without To erase all coagucheek results: Preparing For A Test 1. The correlation was 0.
Page 26 Repeat the test. Open Battery Compartment With the meter turned off, turn it over. Allow to Cozguchek With the cover off, allow the The CoaguChek XS Meter includes advanced features that you can test strip guide to dry for set up as necessary.
Blood 1 hanging drop of blood may be collected directly using a butterfly or a needle and syringe. There is a potential risk of infection. The correlation coefficient is 0.
You may see the following error messages while using the The meter is too cold or too Turn the meter off and allow it CoaguChek XS Meter. Wurster M, Doran T. The letters mem The CoaguChek XS Meter automatically stores up coaguchdk test indicate that you are viewing a results and their dates and times in its memory.
For more information, see re-insert it. To print the manual completely, please, download it. Performing A Test 3. Keep the tube level, and allow it to fill by capillary action. Record Result Record the result according to test strip.