Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||16 October 2013|
|PDF File Size:||1.75 Mb|
|ePub File Size:||12.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
When I gave it a try, though, it worked fine.
Most Active Software Boards: I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
I have not worked with that instrument, but the Amnual web site has the operating and service manual here.
The forums emoticons took over for a bit there. Guess you never know until you try! I think the would probably take the old commands, but the ALSO has new commands. In particular, after setting mxnual and initiating the instrument, I am initiating the measurement and reading it as follows: Message 4 of 6.
Should I be looking at doing it with LabView or something else? Would you mind posting the code you used on the A? Message 6 of 6. Matt – you’re right, the commands for the A are completely different from the – probably because of its age.
Message 5 of 6. You really need the manual.
Keysight / Agilent A – In Stock, We Buy Sell Repair, Price Quote
It’s 84MB, so I hope you have a fast link. However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at. Accepted by topic author Matt C.
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different.
In the code above, MHz is an arbitrary h4p191a.
Thanks for your help guys. I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen. IIRC, the sweep parameters were something like this: I suspect that this difference might be causing my problems.
Message 1 of 6. Sorry, that last line of code there is supposed to read: Message 3 of 6. Message 2 manuao 6. Good luck with it In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
Agilent / HP 4191A RF Impedance Analyzer
If not, what would creating a driver for it entail? Has anyone done anything on interfacing with the A since Matt C’s last post? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.